Pump does not operate
Check Remedy Is the power switch on? Check the switch. Is the power cord plugged in correctly? Check the connections. Is the fuse blown? Call service engineer.
Pump operates but garment will not inflate
Check Remedy Blockage in garment supply tube. Ensure that the tube airway is clear. Garment not fitted correctly to pump. Check for kinked tubing. Air leak in garment. Check garment. Replace if defective.
Needle on status indicator is in yellow zone
Check Remedy Contact Huntleigh Service Department.
Did you know?
Huntleigh Healthcare is one of the world's leading providers of medical diagnostic and treatment equipment since 1979.